Today, the editor wants to share a story with everyone, which is also a story shared by the leader.
This customer previously purchased our grid. The client said on the phone that there was a problem with our X ray grid. The images taken with our grid were black and white on one side and could not be seen at all.
Because we will do a good inspection of the X ray grid before leaving the factory to ensure that the grid is intact and usable. It will be shipped when it is suspected that the grid has been damaged during transportation. The customer sent us The photos of the images are indeed low-density and high-density.
After that, the customer was asked to send us a picture of the grid, and it was found that the customer put the grid upside down. After the client placed the grid correctly, the image returned to normal.
Here are some pictures of the grid:
The editor here must remind everyone that the placement of the grid is positive and negative, we must align the front with the direction of the x-ray, otherwise it will not work
If you need a grid or need to use some grids, you can call us to contact us.
Tel: +86 536-8882360
Company: Weifang Newheek Electronic Tech Co., Ltd.
Address: E Building of Future Star Scientific Innovation Industrial Zone of No.957 Wolong East Street, Yulong Community, Xincheng Sub-District Office, Weifang Hi-tech Zone, Shandong Province, China