The X-ray grid is used to filter out the influence of scattered rays on the film. It is placed between the human body and the film to filter out most of the scattered rays, and only a small part of the scattered rays leak.
X-ray grid classification:
According to the structure:
1 Convergence grid: each lead bar is inclined to the focus, that is, the extension line of the lead bar is concentrated on the focus
2Parallel grid: each lead bar is parallel to each other in the vertical plane
3Cross grid: It is composed of two layers of parallel grids, and the lead strips are 90° to each other.
4Circular arc grid: It is formed by overlapping from the center outwards with equal radius, tile-shaped
According to the material:
1 Commonly used X-ray grid: the internal spacer is a non-metallic material title, and non-metallic low-density materials such as wood chips, paper chips, and plastics are used.
2Full metal X-ray grid: the internal spacer is made of metal material: aluminum sheet, high strength, good moisture resistance
We in Newheek produce and sell various X-ray machines and parts. Products are widely used in domestic and foreign medical and health fields. If you are interested in our products, please contact us.
Tel: +86 536-8882360
Company: Weifang Newheek Electronic Tech Co., Ltd.
Address: E Building of Future Star Scientific Innovation Industrial Zone of No.957 Wolong East Street, Yulong Community, Xincheng Sub-District Office, Weifang Hi-tech Zone, Shandong Province, China